LIFEPAK Service Manual CD-Rom-Service Manual. Physio-Control LifePak AED Service Manual. Physio-Control – LifePak AED by Physio-Control. Physio-Control – LifePak AED. Download PDF. The Physio Control Lifepak AED has been designed to be used by first responders for cardiac arrest Physio-Control LifePak AED Service Manual.
|Published (Last):||20 August 2016|
|PDF File Size:||9.36 Mb|
|ePub File Size:||18.43 Mb|
|Price:||Free* [*Free Regsitration Required]|
Medtronic sevice a registered trademark of Medtronic, Inc. Robotics and Sportster are registered trademarks of U. Microsoft and Windows are registered trademarks of Microsoft Corporation. Troubleshooting During Patient Care Shock Advisory System Appendix B: Figure Test load connection A digital audio recording of scene activity is available as an option. Recorded data may be transferred by direct connection to a printer or computer or by a modem to a remote computer.
Redmond, January mnual, Michael D. Familiarize yourself with all of these terms, warnings, and symbols.
Immediate hazards that will result in serious personal injury or death. Hazards or unsafe practices that could result in serious personal injury or death. Medtronic Physio-Control devices, electrodes, and cables are latex-free. Defibrillation protected, type BF patient connection Attention, consult accompanying documents Warning, high voltage Indicator, steady display indicates battery is low, replace battery; Examine the AED and accessories for any sign of damage during shipping.
Make sure that all the required supplies and accessories, including electrodes and batteries, are present. Save the shipping container and foam inserts for use in reshipping the AED. Steady display indicates the AED requires service by authorized indicator service personnel; flashing indicates service is required immediately.
If the AED is not connected to a patient and no buttons are pressed for 15 minutes, the AED will automatically turn off. The dial string may include up to 20 characters as described in Table The factory default dial string is T9W The characters T9W are required if 9 must be dialed first to access an outside line from the telephone being used. The settings are saved. The actual clinical procedures that you use may vary according to your local protocol.
They are designed for use with devices equipped with the appropriate connector or therapy cable. For more information about these electrodes, refer to the respective electrode operating instructions.
Follow the contour of the ribs and spaces when pressing the electrodes onto the torso.
This limits air space or gaps under the electrodes and promotes good skin contact. Accent marks are not included in message prompts for international languages. You will hear the corresponding voice prompt. Within 10 seconds, the AED displays two alternating messages: The messages alternate without further voice prompts.
If the motion does not cease within 20 seconds, analysis will stop. You must then push if button present to restart analysis. A patient record is created when the AED is connected to a patient and begins to store data.
The Test Log lists the test results and any fault codes detected. The Test Log is printed automatically when data is sent to a printer. Equipment connections for data transfer by modem Figure 5 Turn on the modem. Table summarizes the resources required to print AED data.
Medtronic LIFEPAK 500 Operating Instructions Manual
No entries found Minor Fault Log: For troubleshooting information, refer to page Topics in this section include: If the automatic self test detects a low battery condition or a condition that requires service, the monophasic AED activates an alarm; AEDs with a readiness display will change the indicators servjce the display and will not activate an alarm unless is configured ON.
Confirm that connectors engage Replace damaged or broken parts. Use only the cleaning agents listed in the table. Do not clean any part of the AED or accessories with bleach, bleach dilution, or phenolic compounds. During the self-test, the AED displays the following messages: If the AED requires service, the service indicator appears.
Contact authorized service personnel to perform service. The test will be halted and the AED will operate normally. If the AED detects a problem during the test, the service indicator and message appear. To repeat the test, turn off the AED and then turn it on again. The nonrechargeable lithium battery pak requires less maintenance than the rechargeable SLA battery pak since lifepai never requires recharging. Although a lithium battery may be stored for up to 5 years, it self-discharges over time.
The SLA battery pak should be recharged monthly or after each use. Maintain a battery recharge record.
MEDTRONIC LIFEPAK AED | Physio-Control LIFEPAK | LIFEPAK AED
Recycling Rechargeable Batteries Recycle SLA battery paks locally according to national, regional, and local governmental regulations. If recycling is not possible, contact a Medtronic Physio-Control representative for information or assistance.
In the US, call 1. To promote awareness of battery recycling, SLA battery paks are marked with this label Do not disassemble the defibrillator. It contains no operator serviceable components and dangerous high voltages may be present. Contact authorized service personnel for repair. For mwnual about ordering, contact your local Medtronic Physio-Control representative.
In the USA, call 1. Supplies, Accessories, and Training Tools Table Shelf Servicce 5 years Weight 0.
Specifications apply from 25 to ohms. Voltage compensation is limited to the voltage that would result in delivery of joules into 50 ohms. Therefore, we conclude that biphasic shocks for VF delivered at srevice energy levels have the potential to improve outcome in resuscitation of patients with cardiac arrest. This section also describes screen messages, voice prompts, and event types.
Physio-Control LifePak 500 Service Manual 26500-000036
Press to begin ECG analysis. This system makes it possible for individuals not trained to interpret ECG rhythms to provide potentially-lifesaving therapy to victims of ventricular fibrillation or pulseless ventricular tachycardia.
The results are summarized in Table A If the motion does not cease within 20 seconds, analysis attempts will stop until the operator presses the button again.
Contact your local Medtronic Physio-Control representative for assistance. Recycling of Disposable Electrodes After disposable electrodes are used, follow your local clinical procedures for recycling.
Packaging Packaging should be recycled according to national and local regulations. Page of Go. Page 84 – Table Troubleshooting During Printin Table of Contents Add to my manuals Add. Chapters Table Of Contents Page 6 Troubleshooting During Patient Care Page 7 Figure Test load connection Preface refer to Specifications, page Page 23 Steady display indicates the AED requires service by authorized indicator service personnel; flashing indicates service is required immediately.
Defining Setup Options The dial string may include up to 20 characters as described in Table Page 29 CPR cycle. Page 30 AED data. Aed Operation Follow the contour of the ribs and spaces when pressing the electrodes onto the torso. Page 42 You will hear the corresponding voice prompt.
Page 43 CPR timer. Page 45 20 seconds, analysis will continue. Page 51 Equipment connections for data transfer by modem Figure 5 Turn on the modem. Page 52 13 Turn off the AED and prepare it for the next patient use. Cleaning Confirm that connectors engage Replace damaged or broken parts. Testing During the self-test, the AED displays the following messages: Page 65 The test will be halted and the AED will operate normally.
Battery Maintenance The nonrechargeable lithium battery pak requires less maintenance than the rechargeable SLA battery pak since it never requires recharging.